In this chapter we will cover:
- What the reset and restart options cover.
- How to perform these actions via the Transmitter/display or VPStudio 2.
Factory reset: resets the device back to the factory default settings. This includes deleting all data log sessions and it can be performed via:
Restart: restarts the flow meter. It can be performed via:
3. Adjusting manually via the Transmitter / display
3.1 Factory reset
Go to the Menu >> Factory reset.
Confirm the reset by pressing the up and down button simultaneously.
|Warning: a factory reset will remove all of the recorded data from the device!|
|Go to the Menu >> Restart.|
|Go to the Menu >> Advanced >> Reset .|
Press “Enter” to start to restart the device.
The reset option is also required when updating the display firmware.
4. Adjusting via VPStudio 2 software
Factory reset only, VPFlowScope M only.
In VPStudio 2 unfold the settings menu of the flow meter and select the “Settings” option.
Go to the last tab on the page “Miscellaneous”:
- At Factory reset: press “Reset device to factory settings”.
- Then press “Save”.
A factory reset is not possible via the Internal Web Server, in order to perform a factory reset use the Transmitter or make use of VPStudio 2.
6. Next step
Now that you finished the communications settings the next step is the mechanical installation of the flow meter.